When collecting vibration data with a portable collector, the transducer must be securely mounted to the collection point. I generally use a flat rare earth magnet, but there are some points that are "less than flat". An easy solution is to just pack an extra transducer with a 2 pole magnet in your pocket. It is quick and easy to change transducers, and you will ensure accurate data. Just make sure the transducer sensitivities are the same.
Reader tip provided by Gary Benner Vibration Analyst Georgia-Pacific Camas Washington
Thanks for the tip Gary - your Stainless Steel Diamond Plate Reliabilityweb.com coffee mug is on the way!
Click here to send in your own maintenance tips and get a Maintenance Tip hat or Stainless Steel Diamond Plate Reliabilityweb.com travel coffee mug if we publish it. Link click here to: http://reliabilityweb.formbin.com/forms/tips
“R.A.I.” the Reliability.aiTMChatbot
You can ask "R.A.I." anything about maintenance, reliability, and asset management.